We developed a mixing device that allows Inline mixing of equal volumes of fibrinogen and thrombin. Tank mixing of the two components is not possible as polymerization occurs as soon as they are in contact with each other. Indeed, thrombin catalyzes the conversion of fibrinogen into fibrin and the kinetic rate depends on its concentration - the higher the thrombin concentration used, the faster the conversion into fibrin occurs.
The Inline mixing device is constituted of a plastic molded housing containing 1 or 2 discs of porous material. The disc has a diameter of 3.5 mm, a thickness of 1.5 mm and is made of sintered polypropylene. The disc is characterized by a pore size ranging from 10 µm to 130 um. The housing is equipped with ISO female and male luer ends to allow connection with the ancillaries of the fibrin glue kit. The porous material used can be metal, glass, ceramic or polymers.
The quality of the fibrin material coming out of the mixing device has been evaluated by measuring the percentage of á chains inside the fibrin formed 30 minutes after mixing the fibrinogen and thrombin. we observed that:
1) Mixing with our device is not dependent on the temperature of the two components, respectively fibrinogen and thrombin. The mixing device works as well at 4°C 18°C, 22°C, and 26°C as it does at 37°C.
2) Mixing with our device is not dependent on the low flow rate of the two components, respectively fibrinogen and thrombin. The mixing device works as well at 4ml/min, 8ml/min or 12ml/min. These are the extreme flow rates used by a surgeon to apply fibrin glue in an operating room.
3) A device made of two discs mixes better than with 1 disc made of the same material and in a similar housing. Both are far better than the control. It seems that, for our application, the distance between the two discs plays an important role in the quality of the fibrin obtained.
In our application, ultrasonic welding is used for the assembly of the different parts but they could be press-fitted into a cavity, depending on the device design.
The porous material discs are tool cut but they can be laser or ultrasound cut.
The porous material fulfills the requirements of European pharmacopoeia, USP, JPN. The porous material can be sterilized with gamma, beta irradiation, steam, peracetic, and ethylene oxide, etc.
The porous discs remain static during the mixing.
The design and configuration of the inline mixing device is optimized for mixing fibrinogen and thrombin for a particular surgical application. The dimensions, shape, porosity, material, conditions of use, etc. are not limited to those described in this application.&'
Would you like to access this content?
No problem. You just have to complete the following steps.
You have completed 0 of 2 steps.
You must be logged in to view this content. Log in now.
Purchase Technical Presentation
You must purchase this technical presentation using one of the options below.
If you already purchased this content recently, please click here to refresh the system's record of ownerships.
|Credits||0.5 Use credits|
|List Price||$25.00 Buy now|
|AIChE Members||$15.00 Buy now|
|AIChE Undergraduate Student Members||Free Free access|
|AIChE Graduate Student Members||$15.00 Buy now|